MAX-700B series, MAX-700C series, MAX-FIP, MAX-900 series, MAX-5205
MaxTester battery power level is showing a higher number than 100% while the Battery indicator displays X.
Procedure to follow
- This can happen when you remove a battery that was discharged and replace it with a battery that is fully charged.
- This will not prevent use of the instrument and the instrument will self calibrate during the next battery cycles.
- You can also choose to allow the unit to fully discharge and then to recharge, forcing a battery calibration